[PATCH v4 1/3] dt-bindings: iio: pressure: Support Honeywell mprls0025pa sensor

Andreas Klinger
2023-05-05T13:27:41Z

Honeywell mprls0025pa is a pressure sensor series. There are many
different models with different pressure ranges, units and transfer
functions.

The range and transfer function need to be set up in the dt. Therefore
new properties honeywell,pmin-pascal, honeywell,pmax-pascal,
honeywell,transfer-function are introduced.

Add dt-bindings.

Signed-off-by: Andreas Klinger <ak@it-klinger.de>
---
.../iio/pressure/honeywell,mprls0025pa.yaml | 104 ++++++++++++++++++
1 file changed, 104 insertions(+)
create mode 100644 Documentation/devicetree/bindings/iio/pressure/honeywell,mprls0025pa.yaml

diff --git a/Documentation/devicetree/bindings/iio/pressure/honeywell,mprls0025pa.yaml b/Documentation/devicetree/bindings/iio/pressure/honeywell,mprls0025pa.yaml
new file mode 100644
index 000000000000..84a61721b597
--- /dev/null
+++ b/Documentation/devicetree/bindings/iio/pressure/honeywell,mprls0025pa.yaml
@@ -0,0 +1,104 @@
+# SPDX-License-Identifier: (GPL-2.0-only OR BSD-2-Clause)
+%YAML 1.2
+---
+$id: http://devicetree.org/schemas/iio/pressure/honeywell,mprls0025pa.yaml#
+$schema: http://devicetree.org/meta-schemas/core.yaml#
+
+title: Honeywell mprls0025pa pressure sensor
+
+maintainers:
+ - Andreas Klinger <ak@it-klinger.de>
+
+description: |
+ Honeywell pressure sensor of model mprls0025pa.
+
+ This sensor has an I2C and SPI interface. Only the I2C interface is
+ implemented.
+
+ There are many models with different pressure ranges available. The vendor
+ calls them "mpr series". All of them have the identical programming model and
+ differ in the pressure range, unit and transfer function.
+
+ To support different models one need to specify the pressure range as well as
+ the transfer function. Pressure range needs to be converted from its unit to
+ pascal.
+
+ The transfer function defines the ranges of numerical values delivered by the
+ sensor. The minimal range value stands for the minimum pressure and the
+ maximum value also for the maximum pressure with linear relation inside the
+ range.
+
+ Specifications about the devices can be found at:
+ https://prod-edam.honeywell.com/content/dam/honeywell-edam/sps/siot/en-us/
+ products/sensors/pressure-sensors/board-mount-pressure-sensors/
+ micropressure-mpr-series/documents/
+ sps-siot-mpr-series-datasheet-32332628-ciid-172626.pdf
+
+properties:
+ compatible:
+ const: honeywell,mprls0025pa
+
+ reg:
+ maxItems: 1
+
+ interrupts:
+ maxItems: 1
+
+ reset-gpios:
+ description:
+ Optional GPIO for resetting the device.
+ If not present the device is not resetted during the probe.
+ maxItems: 1
+
+ honeywell,pmin-pascal:
+ description:
+ Minimum pressure value the sensor can measure in pascal.
+ $ref: /schemas/types.yaml#/definitions/uint32
+
+ honeywell,pmax-pascal:
+ description:
+ Maximum pressure value the sensor can measure in pascal.
+ $ref: /schemas/types.yaml#/definitions/uint32
+
+ honeywell,transfer-function:
+ description:
+ Transfer function which defines the range of valid values delivered by the
+ sensor.
+ 1 - A, 10% to 90% of 2^24 (1677722 .. 15099494)
+ 2 - B, 2.5% to 22.5% of 2^24 (419430 .. 3774874)
+ 3 - C, 20% to 80% of 2^24 (3355443 .. 13421773)
+ $ref: /schemas/types.yaml#/definitions/uint32
+
+ vdd-supply:
+ description: provide VDD power to the sensor.
+
+required:
+ - compatible
+ - honeywell,pmax-pascal
+ - honeywell,pmin-pascal
+ - honeywell,transfer-function
+ - reg
+ - vdd-supply
+
+additionalProperties: false
+
+examples:
+ - |
+ #include <dt-bindings/gpio/gpio.h>
+ #include <dt-bindings/interrupt-controller/irq.h>
+ i2c {
+ #address-cells = <1>;
+ #size-cells = <0>;
+
+ pressure@18 {
+ compatible = "honeywell,mprls0025pa";
+ reg = <0x18>;
+ reset-gpios = <&gpio3 19 GPIO_ACTIVE_HIGH>;
+ interrupt-parent = <&gpio3>;
+ interrupts = <21 IRQ_TYPE_EDGE_FALLING>;
+ honeywell,pmin-pascal = <0>;
+ honeywell,pmax-pascal = <172369>;
+ honeywell,transfer-function = <1>;
+ vdd-supply = <&vcc_3v3>;
+ };
+ };
--
2.30.2